Service and Troubleshooting
Other than the Battery Replacement and Reset (as described in this User Manual), the Activforce 2 dynamometer is designed for service by authorized personnel only. For further information, please contact firstname.lastname@example.org.
4435 Eastgate Mall Suite 100
San Diego, CA 92121, USA
Device User Guide
Activforce 2 provides real-time strength and range of motion measurements to quantify patient progress in medical applications, such as physical therapy and muscle rehabilitation. Activforce 2 functions as a handheld device to provide force and angle measurement when connected to an easy-to-use mobile companion application. Download the free Activforce 2 companion application from the Apple App or Google Play store.
Activforce 2 Components
Activforce 2 device
Hand and leg straps
Belt loop plate
Custom carrying case
Attaching a Pad To The Device
Turning The Device On or Off
LED INDICATOR LIGHTS:
A BLUE flashing light indicates that the Activforce 2 device is available for a Bluetooth® connection to the companion mobile application.
A GREEN flashing light indicates the Activforce 2 device is connected to the companion mobile application.
Multiple RED continuous flashing lights indicate that the battery is low and needs replacement.
A single RED flash indicates that the device has been turned off.
Using the Device with the Hand Strap
Proper Usage of the Activforce 2 Device
The Activforce 2 device measures accurately when force is applied in a perpendicular
direction to the large surface areas of the device or an attached pad. Application of the force should be in the center of the surface as much as possible. Direct all force into the device directly perpendicular through the device center point. Follow the instructions provided by the Activforce 2 mobile companion application for best results.
Do not apply force in a non-perpendicular direction to the device.
Do not apply force at an angle or in a direction going across the device.
Do not apply force on or near to the blue band of the device.
Do not apply a force during the three second calibration period before the application starts the force measurement.
Do not move or shake the device during the force measurement.
Measuring Range of Motion
The Activforce 2 device also functions as a way to accurately measure angular motion when used properly. While the device is calibrating, make sure that the device is being held still in the starting position. Then, move the device in a Point A to Point B motion only. Moving the device from Point A to Point B and back to Point A can increase the possibility of an inaccurate measurement. To measure angles accurately, the Activforce 2 device should be rotated on a single axis and kept within a single plane of motion.
Do not rotate the Activforce 2 device in multiple axes during a measurement.
Do not move the Activforce 2 device in multiple planes of motion during a measurement.
Do not rotate the device during the brief calibration period before the application starts the range of motion measurement.
Ensure consistent results over multiple tests by using the same testing techniques and methodology.
Sign in and out of the app or use the security features on your mobile device to ensure data privacy.
Using the Hand Strap
When using the Hand Strap attachment, ensure the strap is securely fixed in position. The Hand Strap may be used to secure the device to the user for strength and range of motion tests.
Using the Soft Pad
When using the Soft Pad attachment, ensure the attachment is securely attached in the locking mechanism. Use on hard or bony body parts for strength tests.
The soft pad can be used with or without the adhesive foam pad and silicon cover. If the adhesive foam pad is used, make sure it is securely fastened to the soft pad and the silicon cover is fully attached.
It is important to apply force directly into the foam pad and not at an angle. This will reduce the risk of the foam tearing off or the silicon cover slipping resulting in inadvertent failure to complete the test.
Utilize the curve of the Soft Pad to place the device against the curved body parts.
Using the Round Pad
When using the Round Pad attachment, ensure it is securely attached in the locking mechanism. Use this pad for all grip measurements. Use this attachment when pushing up against an immovable object.
Using the Belt Loop Plate
When using the Belt Loop Plate attachment, keep the belt positioned perpendicular through the loops to reduce strain on the belt loops.
The plate is designed to hold the belt in position during a test and it is not designed to withstand strain or force from a test. No tension should be applied on the loops for a more accurate measurement. Keep the straps parallel to the direction of applied force so the force is transmitted to the device. Ensure the belt is wrapped over the smooth side of the device and not over the attachment side of the device.
The Round Pad with the Belt Loop Plate is recommended when the device is being pressed into the inanimate object.
The Belt Loop Plate may be used to guide the belt when the device is strapped to a fixed or immobile object.
Using the Leg Strap
When using the Leg Strap attachment, ensure the strap is securely fixed in position. Use the Leg Strap to attach the device to the subject’s larger body parts for range of motion tests.
Cleaning and Disinfecting the Device
To prevent the spread of infection and diseases, the Activforce 2 device and pad should be disinfected between each use. The Activforce 2 device external surfaces, including the rounded pad and the soft, arched pad silicon sleeve, can be hand wiped at room temperature with a soft cloth and a cleaning solution consisting of 70% isopropyl alcohol and water. After application, the cleaning solution should be allowed to evaporate completely.
Alternatively, a cleaning solution of 1% hypochlorite and water (10,000 ppm) can also be used to wipe the device, pads and silicon sleeves. Antibacterial hand sanitizing agents such as non-alcohol wipes or alcohol based hand gels can also be used. Any residue from these agents can be wiped away using a soft, dry clean tissue or towel.
Replacing the Battery
The Activforce 2 device comes with a standard AAA battery, which lasts between six to twelve months, depending on usage and environmental conditions.
To replace the battery, follow these steps.
Obtain a new AAA battery and a flat-head screwdriver.
Hold Activforce 2 device and the screwdriver with the pad attachment mechanism facing upwards. The Activforce logo should NOT be visible.
Insert the tip of the screwdriver in the small opening near the power button and gently turn the screwdriver until a CLICK sound is heard and the bottom shell lid pops slightly open. This opens the first of five latches on the bottom lid.
Gently slide the screwdriver clockwise along the edge of the Activforce 2 device, keeping the tip inside the gap between the lid and the device, until it is about an inch and a half away from the original opening near the power button.
Stop sliding and gently turn the screwdriver until a CLICK sound is heard. Continue sliding the screwdriver around the edge with the tip inside the crevice, stopping every inch and a half to turn the screwdriver to pop open the latches of the bottom lid. The bottom lid can be removed when four to five of the latches are opened.
Warnings and Contraindications
The Activforce 2 device should NOT BE USED when:
The user has a known allergy to silicon or plastic materials.
The skin contacting the device or pad is damaged, irritated or otherwise not healthy.
The device or attached pad is visibly damaged.
The device is in an environment with high heat, humidity or pressure.
This information is provided for general informational purposes only and is not intended as, nor should it be considered a substitute for, professional medical advice. Do not use the information on this website for diagnosing or treating any medical or health condition. If you have or suspect you have a medical problem, promptly contact your professional healthcare provider. The information on this website is provided “as is” without any representations or warranties, express or implied.
If the Activforce 2 device does not connect with the companion application, ensure that the application has Bluetooth© permissions on the mobile device. For some devices, the Activforce 2 companion application also requires Location permissions. If connectivity problems continue to occur, follow the instructions to replace the battery.
While the Activforce 2 mobile application performs a tare calibration operation automatically before each measurement during the calibration phase, the device can also be manually tared. To tare the device, select “Calibrate Your Device” from the settings menu of the Activforce 2 mobile companion application.
To perform a hard reset of the device, remove the battery and re-insert it after 30 seconds. The battery and the removable pads are the only user serviceable items.
For further assistance, complaints or comments, please contact email@example.com.
The Activforce 2 device is certified to be compliant with all required safety standards and regulations, including electro-magnetic compatibility and hazardous materials, for all territories it is distributed in. This includes these standards: Safety IEC 60601-1 and EMC EN 60601-1-2.
The Activforce 2 device is not waterproof and should not be immersed in water or other liquids.
Similar to many electronic devices, the Activforce 2 device contains flammable materials and should not be exposed to flames or other excessive heat sources.
The Activforce 2 device has a service life of at least 30,000 cycles, or force measurements.
In the European Union, the symbol below indicates that the product should not be
disposed of with other household waste. Batteries should be discarded separately, and the device should be deposited at an appropriate facility to enable recycling. Please separate these items from other types of waste and recycle them responsibly, following your local government’s disposal guidelines.
Activforce 2 Device Specifications
Activforce 2 Device Dimensions
Weight: (with AAA battery) 4.8 oz / 137 g
Length: 3.7 in / 9.5 cm
Width: 3.1 in / 7.8 cm
Height: 1.3 in / 3.3 cm
Volume: (approximate) 8.5 in³ / 139 cm³
Activforce 2 Device Functional Specifications
Force Measurement Range: 0 to 200 lb / 0 to 90 kg
Force Measurement Accuracy: ± (1.4 lb + 5% of the applied force)
Range of Motion Accuracy: ±1° for angles measured from 0° to 33°, ±3% of the measured angle when > 33°
System Requirement: Android 8 or higher, iOS 11.4 or higher
Connectivity: Bluetooth© 4.1
Operating Distance: Up to 7 ft / 2 m from mobile device
Battery: 1 AAA battery
Battery life: Approximately 6 months, may vary with usage or environmental factors
Operating Temperature: Range 32° to 95° F / 0° to 35° C
Storage Temperature Range: -13° to 122° F / -25° to 50° C
Operating Humidity Range: 15% to 93% relative humidity
Operating Pressure Range: 20.7 in Hg to 31.3 in Hg, 700 hPa to 1060 hPa
Activforce 2 Case Dimensions
Weight: 29.5 oz / 835 g
Length: 14.2 in / 36.0 cm
Width: 6 in / 15.2 cm
Height: 3.94 in / 10.0 cm
Volume: (approximate) 334 in³ / 5,472 cm³
Activforce 2 Accessories Dimensions
Hand Strap Length: 13.5 in / 34.3 cm
Hand Strap Width: 1.5 in / 3.8 cm
Leg Strap Length: 31.5 in / 80.0 cm
Leg Strap Width: 1.5 in / 3.8 cm
Immobilization Belt Length: 97.6 in / 247.9 cm
Immobilization Belt Width: 1.5 in / 3.81 cm